ENRAF 854 ATG MANUAL PDF
ENRAF-NONIUS ATG Interface. Tank Gauging Protocols Manual. ENRAF INTRODUCTION. The interface between BBI’s Network // Honeywell Enraf ATG Pdf User Manuals. View online or download Honeywell Enraf ATG Installation Manual. Honeywell Enraf series ATG servo gauge is a reliable, versatile and accurate automatic tank level MB; Honeywell Enraf Servo Gauge ATG Manual.
|Published (Last):||12 September 2012|
|PDF File Size:||10.38 Mb|
|ePub File Size:||1.58 Mb|
|Price:||Free* [*Free Regsitration Required]|
The covers should be turned counter-clockwise until the thread Aclicks in place, then turn clockwise until the covers are fully closed. Caution Do not damage the thread of the covers and instrument and keep the thread free of dirt.
The level, shown on the display and transmitted to the host, will be the dipped level and not the actual displacer position. Correct, if necessary temporarily the motor limit high setting MH, in such a way that the gauge will automatically return to this level after a lock-test command LT. The sequence of steps in a procedure may also be important from the point of view of personal safety and prevention of damage; it is therefore advised not to change the sequence of procedure steps or modify any procedure in any other way.
Open the ball valve. During the test, the level dimension and level type on the display are replaced by exclamation marks!!! Perform a repeatability test to ensure the displacer is on the product surface refer to section 4. Maintenance Page 38 5. This device is placed above the motor limit switch high position.
Keep the drum, the outer magnet and the outside of the magnet cap thoroughly clean. Item MZ sets the highest displacer position during a lock test. Be aware that measuring wire is unrolled from the measuring drum. Deviation from any of the prescribed procedures;! If any danger for health, wear a gas mask and take all necessary precautions.
The level, shown on the display and transmitted to the host, will be the dipped level and not the actual displacer position. For verification of the level calibration, refer to section 4. Check motor and manuall encoder excessive oil?
Enraf servo 854ATG install guide
On tank is the nozzle height Repeat this calibration from item BT for the other 3 test weight combinations: Optionally, the instrument can be provided with software for density measurement of the stored product. DC Check whether the drum circumference is correct; if not, change it. Click the file icon to view the file online, or right-click on the file icon and select “Save Target As If not correct, program the correct ,anual area.
The shielding must be grounded in the cable gland on both ends of the cable. The power rating of the external resistor depends on the value of the resistor. When closing, turn the covers counter-clockwise until the thread clicks into place, then turn clockwise.
If applicable close the ball valve and release the pressure gently. In dip mode the displacer is positioned at distance DH above the product and after a time interval DT, the product is dipped.
If the unbalance is still more than 3 grams replace the drum bearings refer to section 5. Grounding shall be performed in accordance with local regulations.
UN Unlock Unlock the gauge. For this purpose the covers of the servo, drum compartment and terminal compartment are fitted with O-rings refer to figure 5. Unlock Cancel the lock test command. This is the highest allowed position for the displacer during normal operation.
Instruction manual series ATG level gauge
This is done by setting item manula to enable. If the reference level value and the gauge reading differs more than the specified value, the calibration procedure as described in section 3.
Remove the drum compartment cover rear cover.
Mind the isolation voltage of the cores in the cable; refer to the International Standard IEC Fix the Allen key screw of the transport bracket. M10 x 25 S 6 main shaft with magnet S 7 key for main shaft S 9 terminal compartment cover S 12 terminal block assembly S 13 8 wire cable bushing S 10 wire cable bushing S 17 GPS printed circuit board S ztg of fuses 10 pc.
If one of the other two interface measurements is selected, the gauge will remain on maunal interface measurement till the default measurement I1 is selected.
Honeywell Enraf ATG Manuals
Page 20 Commissioning 3. Type of item Description Commands These will force the gauge to execute a special task or function. The floating point format is a fixed format; Standard floating point format: